Buy or Sell New HP Defibrillator Codemaster XL Repair HP Defibrillator Codemaster XL from HP and other popular brands. Check out other equipment from Cardiology
After Sale Service
* Weight: 24 lbs. / 10.9kg
Type: Rechargeable, sealed lead-acid battery. 4 Ah, 12 V nominal.
Charge Time: 2 hours to 90% of full capacity. 18 hours to 100% capacity. Repeated charging to less than 100% will reduce the useful life of the battery.
Capacity: 2.5 hours monitoring or 50 full-energy discharges or 1 hour monitoring and recording.
AC input: 100 to 230 VAC +/- 15%, 50/60 Hz.
Battery Indicators: Illuminated LED indicates battery is charging. Low Battery message appears on the monitor.
Waveform: damped sinusoidal.
Output energy (delivered): 2,3,5,7,10,20,30,50,70,100,150,200,300, and 360 joules.
Charge control: push-button on apex paddle and on front panel.
Charge time: less than 5 seconds to 360 joules with a functional battery installed.
Charge done indicator: display indicates available energy, charge-done light on apex paddle, charge done tone.
Synchronizer: SYNC message appears on the monitor and is annoted periodically on the recorder while in synchronous mode. An audible beep sounds with each detected R-wave, while a marker on the monitor and sync mark on the recorder strip indicate the discharge point.
Paddles: standard paddles are anterior / anterior. Adult electrodes (83 cm sq) slide off to expose pediatric electrodes (21 cm sq). Paddle cord is 10 feet (3m). Full range of internal paddles are available.
No Any Documents Uploaded
Write Your Own Review
Disclaimer : All item information on this page has been supplied by seller and PrimedeQ does not guarantee its accuracy.